ColourSpace can be integrated with NEC's MultiSync displays that use the SpectraView engine monitors for accurate calibration.
- ColourSpace Integration
- Closed Loop Profiling
- 3D LUTs
- Inbuilt TPG Operation
The NEC MultiSync PA Series, some P series, and select large screen displays that utilise the SpectraView engine, can be accurately calibrated via 3D LUTs generated using ColourSpace. The following are the only NEC displays with in-built 3D LUT capability:
- PA241W, PA271W, PA301W, P232W, P241W, P242W, P243W, PA231W, MD301C4, PA242W, PA243W, PA272W, PA271Q, PA302W, PA311D, PA322UHD, PA322UHD-2, MD322C8, X841UHD, X841UHD-2, X651UHD, X651UHD-2, X981UHD, X981UHD-2, X551UHD
- (sRGB Displays P404, P484, P554, V404, V484, V554, V404-T, V484-T, V554-T, P654Q, V654Q, C651Q, P754Q, V754Q, C751Q, V864Q, C861Q, V984Q, C981Q)
NEC calibration is controlled via the Hardware Options menu, accessed via the Profiling window.
- If required, make sure the correct drivers are installed
(If necessary, verify via Windows Device Manager)
- Connect the NEC display to the ColourSpace laptop via a USB connection, as well as a direct video connection with valid signal, and start ColourSpace
(If both a USB and HDMI connection are made to the display from the ColourSpace laptop there may be data connection issues, as some NEC displays can be communicated to via the HDMI cable, as well as via USB)
- Use the display OSD to select the Picture Mode you want to assign the 3D LUT to
(Press the up/down buttons when no OSD is shown to select one of the 5 Picture Modes)
- Set the desired set-up options via the Settings menu, as required
- Open a ColourSpace Profiling window, select Hardware Options, and select NEC from the Hardware drop-down list
- From the Serial Port menu select the connected NEC monitor ID
- To make sure there is no existing calibration data active within the display, select Unity Bypass from the Select LUT option, and press Upload
(Or to keep things simple just select Preset #1 if you selected "Picture Mode #1" previously, etc.)
Note: Potentially, when Unity Bypass data is Uploaded all settings will be nullified, not just the relevant LUTs. Should it be desired that settings other than the LUTs remain as set, upload a real Bypass 3D LUT (generated via LUT Tools/LUT Management/New), as that will only null the relevant LUTs, as selected. Alternatively, reset any necessary settings, such as Brightness/Contrast/Black-Level, after the Unity Bypass data Upload.
To perform Brightness/Contrast/Black-Level changes use NEC's own MultiProfiler Software, using the "synchronise" function in MultiProfiler when switching from ColourSpace to MultiProfiler to force the re-reading all of the relevant settings that the ColourSpace Unity Bypass LUT will have changed.
While NEC displays can be calibrated using ColourSpace via a LUT upload, the actual profiling must be performed via a an external patch generator.
Any patch generator can be used, including direct HDMI/DVI/DisplayPort connection from the ColourSpace laptop.
(Possible data connection issues when both a USB and HDMI connection are made to the ColourSpace laptop, defined above.)
After profiling, a Calibration LUT can be generated as normal within ColourSpace LUT Tools.
With the LUT still held within ColourSpace navigate back to the Hardware Options menu, and select the desired LUT from within the Select LUT options, and Upload.