The X-Rite i1Pro 1/i1Pro 2 Spectroradiometers work in combination with ColourSpace to enable accurate calibration to be achieved for both projection as well as monitors via contact and non-contact use.
- ColourSpace Integration
- 10 nm Optical resolution
- 0.2 to 1200 cd/m2
- i1Pro 1 & i1Pro 2
As with all probes, i1Pro 1/2 operation and function is controlled via the Probe Options menu, accessed via the Profiling window.
The i1Pro 1/2 have no variables that are directly associated with the probe, other than Probe Calibration, plus generic probe settings such as Extra Delay Time, Probe Matching, Probe Offset, etc.
Calibration Settings provides generic probe settings, as well as the Probe Calibration option.
Average Low Light
Average Low Light is a probe generic option, that averages multiple readings in low-light situations using an advanced algorithm built into ColourSpace. The algorithm varies based on the probe in use to provide the best possible results.
The Probe Calibration button will save a calibration reading, made with the i1Pro 1/2 placed on its matched calibration plate.
Make sure the serial number on the calibration plate matches the i1pro 1/2 serial number.
Extra Delay Time
Extra Delay sets a time after the patch has been changed before probe measurement starts.
This can deal with delays in the image pipeline.
Probe Matching is really unnecessary with the i1Pro 1/2, as it is a Spectroradiometer.
The options is provided on a Why Not basis...
Probe Offset enables a user to manually enter offset values for different probes/display combinations.
Perceptual Colour Match is a preferable approach.