Any LS result that is not 'correct' shows the profile data is incorrect in some way.
Looking at the info you have provided in the link there are a number of possible issues.
1 - a 33% patch will be too large for an LG OLED
Go with a patch size that is as small as possisble!
2 - a 15 min time difference between when the probe measurements were made for the probe matching to too long. They need to be done back to back.
Warm up the probe with a different USB port if needed.
3 - What do you mean 'Calibration settings: manual', and why 'Burst' - can you not use AIO?
(If you cannot use AIO you have the earlier i1D3 probe, and that 'could' be an issue?)
4 - if the 'Relax' filter is causing issue that suggests you really do have an underlying problem.
I'm downloading the zip file now. Will comment further when I have look through the profiles and LUTs.
Mob Boss at Light Illusion